After a year trying to get the images printed on an Epson 7900 to conform in a consistent manner to what I was seeing on my DELL 30″ LCD screen, I purchased a new NEC Spectra view 3090 monitor at 40% discount. It is a great monitor, but I still was not able to get a consistent monitor calibration, using the Spectraview 5.0 software and the Eye one pro. In fact the best contrast I obtained was 80:1 and this decreased with each further attempt at calibration. Then I noticed that the Eye one pro was slightly warm to the touch. Could the screen be warming the Eye one pro spectrophotometer? I bought a USB powered fan to cool the Eye one pro during the calibration and Walla! The contrast shot up to 400:1, and the monitor calibration became consistent. So it is necessary to air cool the Eye one pro spectrophotometer to get a decent monitor calibration.
Need to air cool Eye one pro when calibrating monitor.
lesadmin · June 26, 2012 · photography · 0 comments
Paris will always be ParisOctober 20, 2015 · 0 comments<h1>Paris will always be Paris (Paris Sera Toujours Paris)</h1> [caption82760Read more
England revisited 34 years on, Part 1/ 3June 12, 2012 · 0 comments<h2>Part 1. Oxford and the Cotswolds.</h2> This was our first extended visit68230Read more
Added International AwardsOctober 27, 2020 · 0 comments<!-- wp:paragraph --> <p>I have added the international awards that my14760Read more
Paris Revisited, Once Again, Part 3/3 ( Highlights)April 23, 2013 · 0 comments<h4></h4> Ambling (flaner) the streets of Paris is a favourite pastime of49880Read more
Sorry, the comment form is closed at this time.